Pressure Measurement PI I URVISH URVISH PI-I
Importance URVISH PI-I Pressure Safety
Achieve desired output URVISH PI-I Pressure
Pressure Pressure = Force/Area Units of force per unit area URVISH PI-I
Hydrostatic Pressure Hydrostatic pressure is the pressure in a liquid. The pressure increases as the depth in a liquid increases, due to its weight.
In term of equation, P = ghgh = density in kg/m3 g = acceleration due to gravity (9.8m/s2) h = depth in liquid in m
P = pressure in Pa Manometers
U tube Well Type Inclined Float Ring Balance Barometer
URVISH PI-I Manometers Barometer
URVISH PI-I Float type Mano Meter URVISH
PI-I Ring Balance Type Manometer URVISH PI-I
Types of Bourdon Tube C Type Spiral Helical URVISH
PI-I Bourdon tube URVISH PI-I
Spiral Bourdon element URVISH PI-I Bellows
URVISH PI-I Beam balance bellows sensor URVISH
PI-I Diaphragm Type URVISH PI-I
URVISH PI-I Actual Image of Diaphram URVISH
PI-I Electrical Pressure Sensors
LVDT type Strain gauge Piezo electric type Capacitance type. URVISH
PI-I LVDT type URVISH PI-I
LVDT type URVISH PI-I LVDT with Bellows
URVISH PI-I Strain gage Resistance URVISH
PI-I URVISH PI-I Strain gage
Strain gage is attached to a flexible membrane. When the membrane is deflected the resistance in the strain gage changes. URVISH PI-I
Strain gauge with diaphragm URVISH PI-I URVISH
PI-I Piezo electric Piezoelectric Effect is the ability of certain materials to generate an electric charge in response to applied mechanical stress.
URVISH PI-I Piezo electric type Sensor Principles:
Pressure Charge Voltage Applications: measurements with high accuracy, good repeatability, high resolution. URVISH PI-I
Piezo electric Material
Quartz Cane sugar Rochelle salt Topaz Tourmaline Bone
URVISH PI-I Capacitance type URVISH
PI-I Working URVISH PI-I
Wein Bridge URVISH PI-I Actual Transmitter (DPT) Differential Pressure Transmitter
URVISH PI-I Vacuum Sensors
Thermal Conductivity gauge Pirani Gauge Ionization Gauge McLeods gauge
URVISH PI-I URVISH PI-I
URVISH PI-I URVISH PI-I
Pirani Gauge URVISH PI-I Pirani Gauge
URVISH PI-I McLeods gauge MOST IMP
URVISH PI-I URVISH PI-I
McLeod Gauge Formula P= KHH0(1-KH) URVISH PI-I P= measured
Pressure K = Constant H= Height difference in open column H0= Height above zero line closed column
Ionization Gauge Electrodes URVISH PI-I
Ionization Gauge It can be used to measure pressure down to about 2 torr. The gas is ionized with a beam of electrons and the current is measured between two electrodes in the gas. The current is proportional to the number of ions per unit volume, which also proportional to the gas pressure.
Optical Pressure Sensor URVISH PI-I Pressure Switch
NO / NC on SET Pressure URVISH PI-I Pressure Switch
URVISH PI-I URVISH PI-I
Dead weight tester Application : To calibrate Pressure Gauges URVISH PI-I
Dead weight tester URVISH PI-I URVISH
PI-I