NSLS Bending Magnet Photon Energy [keV] X13 Undulator Source The Mini-Gap Undulator [MGU] the brightest source of hard x-rays (E ~ 3.7 keV 16 keV) at the NSLS. MGU parameters: 54 pole, 1.25 cm period At a gap of 3.3 mm , Magnetic field = 0.92 Tesla and Deflection Parameter, K~1.07 Fundamental @ 3.7 keV, 4x1017 ph/sec/mm2/mrad2/0.1%BW/300mA
Brightness (ph/sec/mm2/mrad2/0.1%BW/300mA) MGU Brightness Magnetic Gap=3.3 mm 8x1017 Calculated Experiment 6x1017 4x1017 2x1017
Current X13B Beamline Status monochromator hutch 27 24 20 15
12 numbers (in meters )indicate distance from center of X13 straight-section, where MGU is located. Specifications The X13B microdiffraction instrument will be modular and compact in design, which will make available a variety of x-ray optics for the experimentalist. Focused X-ray Beams on the micron-length scale and below, together with low and high x-ray divergence will be available with these optics.These will be mounted on a motorized rails and will be readily accessible. Pink Beam (Laue) and Monochromatic modes of operation will be available.- Energy tunability will be accomplished through a
25 26 Distance from MGU source [meters] 27 Suite of Optics offered Pin Holes Capillaries KB Mirrors Zone Plates Planar Refractive Lenses (Current R&D)
Kirkpatrick-Baez Elliptical Mirrors Key Features: Achromatic - Energy independent focusingpink beam and monochromatic modes Accurate Elliptical Figures Achievable (either by bending or metal deposition ) Small Spot Sizes Heavy metal Polished spherical Silicon substrate High Reflectivity at Grazing Incidence Angles X13B Microprobe
differentially-deposited elliptical KB optics have been used at X13B during the past few years, designed primarily for high-pressure x-ray spectroscopy. Vertical blade scan current derivative Horizontal blade scan current derivative ~ 9 microns FWHM ~ 3 microns FWHM blade position [ microns ] blade position [ microns ] X13B Microprobe
Calcium K Fluorescence Mapping of Osteoarthritic Bone 1. 3.7 mm 3. 2. 3.3 mm 1 . 2.
3. Preliminary KB layout 25 m to MGU source 2cm incident x-rays focal position 3cm 10 cm Expected Parameters 1 micro-radian figure error Spot size 0.7 x 2 micron2
Divergence 1mrad (v) x 7 mrad(h) 8 keV, 6x109 ph/sec Gain 5x104 With Figure Error (1 micro-radian) X13B Microprobe R&D Efforts -Planar Refractive Lenses Intensity [arb. Unit] (K. Evans-Lutterodt, J.M. Ablett) Position [mm] Summary
A state-of-the-art hard x-ray microdiffraction instrument is proposed for NSLS mini-gap undulator beamline X13B. It will offer a variety of x-ray microfocusing optics micron scale spatial resolution and below & high and low divergence. Big-beam (unfocused ) mode. Pink Beam and Monochromatic Modes of Operation. Simultaneous fluorescence and diffraction information.
Calculated rad-con equilibrium T vs. observed T Zonally averaged net radiation Roles of atmosphere and ocean Basic dynamical relationships Basic dynamical relationships Basic dynamical relationships Geostrophic balance Geostrophic balance Rotating vs. nonrotating fluids Rotating vs. nonrotating fluids Rotating vs. nonrotating...
The HSC minimum standard is a practical level of reading, writing and numeracy used in everyday life, for work and further study . From 2020, students in NSW must meet a minimum standard of literacy and numeracy to be eligible...
What to look for in good writing. THE SIX TRAITS SIX TRAITS IDEAS ORGANIZATION SENTENCE FLUENCY CONVENTIONS VOICE WORD CHOICE IDEAS makes sense narrow, manageable topic sounds like writer knows what he's talking about fresh spin important details ORGANIZATION hook...